FLUKE 80T-IR PDF
80T-IR – read user manual online or download in PDF format. Pages in total: Find great deals for Fluke 80t-ir Temperature Probe 80TIR. Shop with confidence on eBay!. Find great deals for Fluke Infrared Temperature Probe M# 80t-ir. Shop with confidence on eBay!.
|Published (Last):||13 January 2013|
|PDF File Size:||4.69 Mb|
|ePub File Size:||15.61 Mb|
|Price:||Free* [*Free Regsitration Required]|
Save 80t-ri item to a new parts list. Please enter a message. Power up the UUT and allow it to stabilize for five minutes. Add to a parts list. If your instrument fails to pass any performance specifications, return it to the Eluke Eactory or Service Center for repair.
With the UUT powered on, check that the battery voltage is at least 7.
Fluke | 80T-IR TEMP PROBE R | | RS Malta Online
Save to parts list Save to parts list. Verification Points Model Blackbody Temperature at 0.
Search the history of over billion web pages on the Internet. The foregoing information relates to product sold on, or after, the date shown below.
Sending feedback, please wait Accuracy figures are valid for a period of one year. Reactivate by switching the UUT off and on again. In between measurements move the UUT to an environment not affected by the heat source.
May substitute less accurate blackbody if using in conjunction with transfer standard of equivalent 80tt-ir, i. Set the DMM to the 3.
Thank you for your feedback. Replace the battery if necessary. Save to an existing parts list Save to a new parts list. Record the reading displayed on the DMM.
Please select an existing parts list. You have chosen to save the following item to a parts list:. Refer to Figure 1 for equipment configuration.
80T-IR TEMP PROBE R
The product does not contain any of the restricted substances in concentrations and applications banned by the Directive, and for components, the product 80t-it capable of being worked on at the higher temperatures required by lead—free soldering.
Recommended Equipment Table 1 lists the equipment required to perform the calibration verification procedure. The Manufacturers reserve the right to change this Information at any time without notice. Repeat 880t-ir 1 through 5 for the remaining verification points called out in Table 2.
Adjust the setpoint of the Blackbody to output each of the temperatures in 80g-ir 2 at 0. Place the UUT aperature three inches from the center and perpendicular to both axes of the Blackbody plate. You agree that the Information as provided here through an intermediary may not be error-free, accurate or up-to-date and that it is not advice.
The product detailed below complies with the specifications published by RS Components. The Manufacturers disclaim all warranties including implied warranties of merchantability or fitness for a particular purpose and are not liable for 80tt-ir damages arising from your use of or your inability to use the Information downloaded from this website.